Gas Chromatography Mass Spectrometry (GC/MS)
The GC-MS is composed of two major building blocks: the gas chromatograph and the mass spectrometer. The gas chromatograph utilizes a capillary column which depends on the column's dimensions (length, diameter, film thickness) as well as the phase properties (e.g. 5% phenyl polysiloxane). The difference in the chemical properties between different molecules in a mixture will separate the molecules as the sample travels the length of the column. The molecules take different amounts of time (called the retention time) to come out of (elute from) the gas chromatograph, and this allows the mass spectrometer downstream to capture, ionize, accelerate, deflect, and detect the ionized molecules separately. The mass spectrometer does this by breaking each molecule into ionized fragments and detecting these fragments using their mass to charge ratio.
These two components, used together, allow a much finer degree of substance identification than either unit used separately. Combining the two processes reduces the possibility of error, as it is extremely unlikely that two different molecules will behave in the same way in both a gas chromatograph and a mass spectrometer. Therefore, when an identifying mass spectrum appears at a characteristic retention time in a GC-MS analysis, it typically lends to increased certainty that the analyte of interest is in the sample.
May 13. 2013 Smiths Detection’s eqo people screener wins key EU approval [more]
May 08. 2013 Smiths Detection X-ray Customers Offered Real-Time 24/7 Contact with MSA Security Certified Bomb Technicians [more]
April 11. 2013 Smiths Detection Supplies Explosives Detectors to Canadian Airports [more]
April 08. 2013 Smiths Detection Opens New Training and Service Center for the Americas [more]
April 04. 2013 Luxembourg Customs bolstered by Smiths Detection Cargo Scanner [more]
November 26. 2012 Smiths Detection launches specialist X-Ray scanner for cars [more]